The analyser is a fixed point per octave analyser, rather than an FFT linear analyser, allowing greater resolution at low frequencies:
Use the Show Max button to display a set of crosshairs at the maximum amplitude detected.
Use the Show User button to display a set of crosshairs at a user defined frequency. Tap the graph or use the encoder beneath the screen (in Follow Detail Mode) to set the frequency.
The R-L button is only available in the stereo version of the module and shows the difference between Left and Right channels. These can be routed to two separate signals for comparison.
The Function generator provides an oscillator of variable frequency, with a fixed option of 1 kHz, in addition to white noise and pink noise generation:
Use the Frequency Weighting buttons to choose the weighting:
Use the Time Weighting buttons to choose the attack time constant:
The SPL Meter has three Calibration Modes. Each has a varying level of complexity to configure and is proportional to its accuracy.
In Uncalibrated mode the meter will read 140 dB at 0 dBFS. This is the least accurate mode.
Auto mode uses a simple calibration routine to equate a known dB SPL level to a measured dB FS level. Connect a known dB SPL source to the console and route it to an Input Channel. Insert the SPL Meter into the Channel's Insert point and set the Calibration Mode to Auto. Press the Calibrate (Z, Fast) button. This will automatically set the meter's frequency and time weighting to dB(Z) and Fast respectively. Next, enter the known dB SPL level into the Calibration Level field. Once the Equivalent FS level has settled, press the Calibrate (Z, Fast) button again. Do not change the Calibration Mode as this will reset the calibration.
Sensitivity mode requires information about the measurement microphone being used, specifically its sensitivity in mV/Pa. This can usually be found in the specifications section of your microphone's manual or web page.
Route the measurement microphone to an Input Channel and route the Channel's Insert send to a console output. Set the Input Channel's Analogue Gain to a nominal value so that the input is not clipping and set the Digital Trim to 0dB.
Send an input signal of known level (A) into the channel and adjust the console's output trim (in the) until the output level (B) matches the known input level (you may need to adjust the input gain to achieve this).
Do not adjust the gain structure between the input and Insert point from now on as this will invalidate the calibration procedure.
Unroute the console output from the Channel's Insert point and insert the SPL Meter in its place. Set the Calibration Mode to Sensitivity and enter the microphone's sensitivity into the Mic Sensitivity field.
Finally, add the console's operating level (found in Menu > Setup > System) to the output trim offset that was applied to match the input and output levels and enter it into the Output Gain field of the SPL Meter module.
Revert the output trim to your operational requirements and ensure that the gain structure between mic and SPL Meter remains constant (e.g. you may wish to exclude this channel from automation recall scope).
The stereo and LCR Phase Scope modules allow the phase of two or three sources to be compared in real-time.
Type: Set Peak or RMS sensing.
Hold: Set the detector hold time to 3seconds, 10seconds, Infinite or None. There is also an option to RESET the hold indicator.
Reference: Set the reference measurement level.
Integration: Set integration time to 1seconds, 2seconds, 10seconds, 30seconds, 1minute, 2minutes or Infinite. There is also an option to RESET the integration time.
The Monitoring modules are designed to be inserted into Solo Channels or other output channels to provide monitor source compatibility and confidence checking.
The Stereo Monitoring Module provides all the expected compatibility monitoring modes needed for stereo production.
The Multi Format Monitoring Module is a 4.0 or 5.1 Module that includes all the features of the Stereo Monitoring Module but with the additional elements that surround monitoring requires.
Note: As the Solo Channels include EQ, delay and All Pass filters, and with the option of an FX Rack graphic EQ in the other insert point, it's possible to fine tune a surround monitor setup without resorting to outboard processing or other external speaker management hardware.